So far we already said that the DA-meter is an innovative tool for agriculture: it measures the fruit’s ΔA index that is an indicator of its maturation level.
However how do you actually make a measurement with the DA-meter?
Let’s go through the few simple actions you have to make every time you will start a set of measurements with the DA-meter.
1►Turning on the instrument
At this point, unless you did it little before, the display will show a warning message inviting you to calibrate the instrument.
Calibration is a very important process to make accurate measurements, you must not skip this step.
For the calibration process you will need the reference white, that is the white round piece of plastic.
First of all open the main menu with the menu button.
When you enter the main menu the pointer (►) will be already on the first item “CAL” which allows you to start the calibration process. Press the OK button✓ to select it.
Now you have to put the sensor of the DA-meter in contact with the reference white and press OK button✓ again.
You will see a red light around the sensor and you will hear a prolonged ‘beep’. Now the calibration process is completed.
If the process has been correctly performed, the instrument will display the main menu again.
In the event of a calibration failure, it has to be repeated.
3►Clearing the average ΔA data story
Now going through the other menu items the first one after “CAL” is “CONFIG” which allows you to enter the configuration menu that enables to change the personal settings according to the different way of use, but we will examine the various options later on this blog.
Next item we need to select on the menu is “MEAN CLR”.
Later on this blog we will examine in depth the use of the average ΔA data.
At the moment you just need to know that this step is important in order not to falsify the data collection. It has to be done every time you start a new set of measurements in order not to mix previous data with the current ones.
4►Setting to zero the counter of measurements
Now move again the pointer (►) with the down arrow button on the item “CNT CLR” which enables you to set the counter of measurements to zero. Press the OK button✓ to select it.
The instrument will display a message saying the count has been set to zero.
5►Setting a marker for the current set of measurements
Now move the pointer (►) with the down arrow button on the next item on the main menu which is “SET MARKER” and press the OK button✓ to select it.
The instrument will show you a number, which is the new marker, that is a serial figure to identify the current set of measurements.
6►Measuring the ΔA index
To make a measurement you just have to put the sensor of the instrument in touch with the fruit and press the OK button✓. You will see a red light around the sensor and you will hear a prolonged ‘beep’.
The ΔA index has been measured and it will show in the black strip on the lower part of the display.
It is important to know that the instrument includes various data collecting options that could be changed when needed from the configuration menu. It is important to distinguish the options in order always to select the more suitable one.
We are going to examine them in a next post.